MOQ : 50 Piece
Business Type : Exporter, Supplier
|Nominal width||DN 10 (RRH-010) , DN 25 (RRH-025)|
|Pressure resistance||PN 100 bar|
|Medium temperature||0..100 °C|
|Current consumption||30 mA|
|Output current||max. 100 mA|
|Electrical connection||cable 2 m or for Round plug connector M12x1, 4-pole|
|Resistant to short circuits||yes|
|Reversal polarity protected||yes|
|Ingress protection||IP 67|
Preferred Buyer From
Flow Transmitter RRH :
Characteristics : The flow meter consists of a spinner which is rotated by the flowing medium. The rotor's rotational speed is proportional to the flow volume per unit time. The rotor is fitted with magnets. A Hall sensor records the rotational speed, which is proportional to the flow rate.
Wiring : Air bubbles affect the measurement results. For filling processes, the valve should be installed behind the sensor. A running up time of approx. 0.5 seconds and a running down time of approx. 3 seconds should be noted.
Handling and operation :
The Rototron device is installed in the pipework with the aid of the rotatable adapter pieces. If necessary, the adapters can be removed from the body of the housing after the stainless steel clips have been removed from the housing. Before reinstalling, it should be ensured that both the adapter with the O-ring and the sealing surface in the body are clean and undamaged. The adapters should be fitted carefully in the housing (it is best to turn them), so that the O-ring is not damaged. With this flow sensor, there is no need for run-in and run-out sections. However, it should be ensured that the flow sensor is at all times filled with medium. Any preferred installation position is possible, but the best possible venting position should be chosen (rotor axis horizontal, flow horizontal or from bottom to top).